Welcome to FindLaw's Cases & Codes, a free source of state and federal court opinions, state laws, and the United States Code. For more information about the legal concepts addressed by these cases and statutes, visit FindLaw's Learn About the Law.
(a) Identification. A balloon aortic valvuloplasty catheter is a catheter with a balloon at the distal end of the shaft, which is intended to treat stenosis in the aortic valve when the balloon is expanded.
(b) Classification. Class II (special controls). The special controls for this device are:
(1) The device must be demonstrated to be biocompatible.
(2) Sterility and shelf life testing must demonstrate the sterility of patient-contacting components and the shelf life of these components.
(3) Non-clinical performance evaluation must demonstrate that the device performs as intended under anticipated conditions of use, including device delivery, inflation, deflation, and removal.
(4) In vivo evaluation of the device must demonstrate device performance, including the ability of the device to treat aortic stenosis.
(5) Labeling must include a detailed summary of the device-related and procedure-related complications pertinent to the use of the device.
Cite this article: FindLaw.com - Code of Federal Regulations Title 21. Food and Drugs § 21.870.1255 Balloon aortic valvuloplasty catheter - last updated October 03, 2022 | https://codes.findlaw.com/cfr/title-21-food-and-drugs/cfr-sect-21-870-1255.html
FindLaw Codes may not reflect the most recent version of the law in your jurisdiction. Please verify the status of the code you are researching with the state legislature or via Westlaw before relying on it for your legal needs.
Was this helpful?